A large back-lighting LCD makes the screen clear and easy to read. Quick processing for measurement data of 18 items including 3 differentials of WBC and for 3 items of histogram only by small volume of sample suction. Measurement of 18 parameters and histograms are quickly processed within 60 2. The two types operation of swing and roll make mix easy satisfactorily.
|Published (Last):||8 November 2013|
|PDF File Size:||20.53 Mb|
|ePub File Size:||18.86 Mb|
|Price:||Free* [*Free Regsitration Required]|
So, please be sure to read carefully. Symbols and their meaning are as follows. Those conducts may risk fire and electrical shock. The use of Power Unit other than designated here might risk fire and electrical shock. Power Unit outlet should be independent. If and outlet is shared with other electric devices, the outlet heated extraordinarily and is subject to a cause of fire.
Pull a plug out by holding a plug with two fingers, or it may Cause electrical shock or short to bring about fire. Not to plug or unplug with wet hands. It may cause electrical shock. Not to damage power cords and connecting cables, etc. Do not step on, twist or pull cords and cables. Power cords and connecting cables get broken and this may. Not to use damaged power cords and connecting cables, etc.
Never use other cords and cables than the designated ones. Cords and cables with less electrical capacity may cause fire. Stop operation when in extraordinary motion. Shut off power and unplug a cord immediately when you. Take Ground. Unless it is done, it may cause an electrical shock when breakdown and electrical power leaking occur. CAUTION Do not use thinner or benzene as it may cause change of color or shape for external stains of the unit, wipe with soft cloth or cloth slightly wet.
For heavy stains, wipe with diluted detergent solution or with alcohol. Do not put foreign matters such as bolts or coins inside the unit. It may cause breakdown. Use and operate the unit in accordance with enclosed "User's Guide" Handle switch keys of the Touch Panel with hand care.
Rough handling keys and handling them with a tip of sensor driver or ball. For operation of switches and keys, do it one by one Doing two or more switches at the same time may cause breakdown or damage. Touch panel LCD display Displays measuring results, messages and operation status.
Menu setting can be done by touching the panel. Pilot light While this light is on, power of the unit is on, even when LCD back light is off. Shutting off the unit can be normally done by touch panel. However, in case of emergency, this switch also shuts off the unit, however, cleaning is not to be done.
Reagent holder panel At the time of changing reagents, Hemolyzer bottle, Cleaning solution bottle are to be changed. After suction, the probe draws back in the unit for dilution. START switch 3. The switch starts suction of sample for measurement. The switch works only when suction of sample is so requested and possible. Printer cover The cover is to be opened for changing print paper.
Paper cutter This cutter is for print paper use. Paper can be cut by pressing the paper downward or upward. Every time the button is pressed, the printer prints out. FEED: While the button is pressed, print paper keeps coming out. This is to be connected with PC for data output.
Connecter of external printer External Printer is to be connected. Cooling fan It keeps the temperature in the unit stable. It operates all the time while power to the unit is on. Fuse holder A fuse is held in.
AC inlet To be connected with the attached power cable and the other side to be connected to AC outlet. Earth terminal In case 3-pin type outlet can not be applied, or 2-pin type outlet has no terminal, take Earth with this terminal.
Water supply joint To be connected to water supply tube Blue mark. Cleaning solution joint To be connected to Cleaning solution tube Green mark. Detector maintenance panel Open the panel when cleaning detector. Heamolysis reagent nozzle There is a hemolysis reagent nozzle with cap in the holding space for aspirating the Hemolysis reagent.
When the Hemolysis reagent is changed, exchange the Hemolysis reagent bottle for new one removing the cap. Roller pump-XB1 This pump is for washing the sample aspiration nozzle. Change the tubing for rollar pump periodically. Roller pump-XB2 This pump is for aspirating the Diluent. Roller pump-XB3 This pump is for discharging the waste solution.
Pinch valve XB4 This valve is for changing the discharge from waste solution tank or counting chamber. Take PCE unit and its accessories neatly out of cartons.
By reference to the check list, make sure that no accessories are missing and no damages. If any, contact your local agent immediately. Set PCE unit on a stable working table in an inspection room. Refer to Installation procedure. Use the unit in the environment where room temperature is in a range of Particularly, make sure that room temperature for a diluent will not go out of range to secure stable results.
Install the unit on a flat and sturdy table that is not subject to vibration. For installation, avoid the place that is subject to direct sun, high humidity and water-splashing near a sink etc. Do not put a container containing reagent, water, etc. If the unit is drenched, it may cause breakdown. Do not use the unit in a dusty place, as it is a precision instrument to measure micro particles. Electricity consumption is approx.
Do not use the same power outlet that is also used for airconditioner, refrigerator, micro-wave cleaning instrument and constanttemperature bath etc. Also, keep the unit about 24m away from those units. Be sure to use an independent power source for the unit. Particularly, use the other wall outlet for devices that generate noise. Put a connector of the attached power cable into AC outlet on the back panel, then push a plug at the other end of the cable in wall outlet. The end of a plug is equipped with earth terminal.
In case that ground is not obtained by the terminal, be sure to apply ground terminal on the rear panel of the unit for securing grounding. Be careful not to let ground tube bend over. Try to put a diluent solution bottle on the same height as the unit, as much as possible. Try to put a waste solution bottle on the same height as the unit, as much as possible. Take off rubber cover over a joint on the right side panel.
Connection of diluent bottle Connect a diluent bottle tube to a joint blue on the right side panel, then connect the other end of the tube to the joint of a diluent bottle. Connection of Cleaning solution bottle Connect the reagent nozzle tube green to the joint green in the reagent holder inside Front panel, then connect the other end of the nozzle tube to a Cleaning solution bottle. Connection of waste solution bottle Connect a waste solution bottle tube to a joint red on the right side panel, then connect the other end of the tube to the joint of a waste solution bottle.
Connection of float sensor Connect float sensor cable to the connector on the right side panel. Connection of Hemolyzer bottle Connect the reagent nozzle tube yellow to the joint yellow in the reagent holder inside Front panel, then connect the other end of the nozzle tube to a Hemolysis Pull out the ratch of printer cover toward you while pushing printer cover stopper downward.
Printer cover can be removed. Paper cutter. Pull printer lever down toward you. Insert the top end of a printer roll into the slot of print paper insertion. Paper holding lever. After print paper comes out of print head, put printer lever back to its position.
When you do so, make sure that print paper is set straight. Paper holding. Pull the end of print paper out of print paper outlet, then push the printer cover back in while making sure that print paper will not get caught. For setting printer cover, put the edge of down part of printer cover in the notch, then push the upper part in. Selfdiagonotic of water-supply and circuit automatically begins.
So, please be sure to read carefully. Symbols and their meaning are as follows. Those conducts may risk fire and electrical shock. The use of Power Unit other than designated here might risk fire and electrical shock.